Plan Your GyroStim Space
We recommend a minimum room size of 12'x14' to comfortably accommodate the GyroStim and operator station.
We will schedule a floor plan review once you have filled out the floor plan template below and shared it with us.
Outline your room dimensions on the template.
Sketch in important item locations:
windows and light switches;
location of 220V & 110V outlets;
room doors and door swing;
permanent architectural features;
equipment and furniture including the operator station.
Place Your GyroStim
The GyroStim has an 8' diameter footprint; sketch your desired GyroStim door swing.
GyroStim Installation Requirements
PATHWAY between delivery and installation point
Provide a video of this path from delivery to GyroStim room.
DOORWAY and HALLWAY DIMENSIONS
Confirm and provide us with the dimensions (W-H) of all hallways and doors along the installation pathway.
For a standard installation, passage of 5’ wide is required,
For a custom installation, passage of 36” is required.
Pathway must be free of steps or stairs.
ELEVATOR for installation not on ground floor
Confirm and provide us with the dimensions (L-W-H) of the elevator if applicable.
Elevator must be a minimum of 3’ x 4’ x 7’
Forklift and trained operator should be available to offload crates from freight truck if required.
A room size of 12 ft. x 14 ft. or larger is recommended.
See floor plan template below.
Confirm floor load bearing capacity from landlord.
Must be able to support 1800lbs distributed over an 8x8 footprint.
Provide Input power of 220VAC 30A for GyroStim
Provide permanent connection box with On/Off disconnect switch, lockable switch:
30 Amp 240-Volt 2-Pole Fused Indoor Light Duty Safety Switch for H or J Fuse Types
Manufacturer: Square D, Model #L221N
Provide additional power of 110VAC for operator station
Two professionally trained movers with dollies and other basic moving equipment should be available to assist with installation if required.